|Year : 2019 | Volume
| Issue : 8 | Page : 1091-1098
Accuracy of CBCT images in the volumetric assessment of residual root canal filling material: Effect of voxel size
F Yilmaz1, G Sonmez2, K Kamburoglu2, C Koc3, M Ocak4, HH Celik4
1 Department of Endodontics, Faculty of Dentistry, Ankara University, Ankara, Turkey
2 Department of Dentomaxillofacial Radiology, Faculty of Dentistry, Ankara University, Ankara, Turkey
3 Department of Endodontics, Faculty of Dentistry, Başkent University, Ankara, Turkey
4 Department of Anatomy, Faculty of Medicine, Hacettepe University, Ankara, Turkey
|Date of Acceptance||04-Apr-2019|
|Date of Web Publication||14-Aug-2019|
Dr. F Yilmaz
Department of Endodontics, Faculty of Dentistry, Ankara University, Ankara
Source of Support: None, Conflict of Interest: None
| Abstract|| |
Aims: Our aim was to compare three different voxel sizes of CBCT images for the determination of residual filling material volume in root canals when compared with micro CT. Material and Methods: Forty-two root canals of 14 extracted human maxillary molar teeth were retreated by using ProFile® instruments. Images were obtained after retreatment by using ProMax 3D Max CBCT at 3 different voxel sizes (1) High resolution (0.1 mm); (2) High definition (0.15 mm); and (3) Normal resolution (0.2 mm). Two observers measured volumes of residual filling materials in exported CBCT images by means of 3D Doctor Software. Micro CT measurements were served as gold standard. Mann-Whitney U test and Wilcoxon Test were used for the comparison of CBCT and micro CT measurements. Statistical significance was set at P < 0.05. Results: No statistically differences were found between the two observers for all measurements (P > 0.05). There were no significant differences among different CBCT voxel sizes used (0.1 mm, 0.15 mm, and 0.2 mm) (P > 0.05). The Spearman correlation coefficients between CBCT at different voxel sizes significantly highly correlated with micro CT measurements for each observer (P < 0.05). Furthermore, no significant differences were found between the measurements obtained by the two observers in consideration to root canal location (P > 0.05). Conclusion: CBCT images may provide useful information in the volumetric assessment of the amount of residual filling material in root canals for retreatment procedures.
Keywords: CBCT, Micro CT, retreatment, voxel size
|How to cite this article:|
Yilmaz F, Sonmez G, Kamburoglu K, Koc C, Ocak M, Celik H H. Accuracy of CBCT images in the volumetric assessment of residual root canal filling material: Effect of voxel size. Niger J Clin Pract 2019;22:1091-8
|How to cite this URL:|
Yilmaz F, Sonmez G, Kamburoglu K, Koc C, Ocak M, Celik H H. Accuracy of CBCT images in the volumetric assessment of residual root canal filling material: Effect of voxel size. Niger J Clin Pract [serial online] 2019 [cited 2020 Jun 5];22:1091-8. Available from: http://www.njcponline.com/text.asp?2019/22/8/1091/264424
| Introduction|| |
Root canal retreatment is indicated in order to eliminate or reduce intracanal and periradicular microbial infection. This treatment is usually considered to be the first treatment option in case of endodontic failure. The retreatment procedure involves complete removal of the previous filling material from the canal system to allow effective cleaning, shaping, and refilling of the root canal system. Several techniques have been proposed for retreatment procedures such as; hand files, NiTi rotary files, solvents, heat, ultrasonics, and lasers. However, none of those techniques were found to be successful for completely removing the whole root canal filling material.,,, Various methods have been used in endodontics in an attempt to evaluate the efficacy of root filling material removal including intraoral periapical radiography, stereomicroscopy and digital camera. However, none of those methods are able to provide quantitative volumetric information before and after conducting retreatment procedures.,,, Assessment of residual filling material can be difficult with periapical radiography due to anatomic noise from surrounding structures, superimpositions, and projection geometry issues. Cone beam computed tomography (CBCT) has been specifically designed to reproduce undistorted 3D information of the maxillofacial skeleton, including the teeth and their surrounding tissues, with a significantly lower effective radiation dose compared with medical computed tomography. A significant limitation of CBCT imaging is the presence of metallic artifacts, which are caused by metal and amalgam restorations and, to a lesser extent, root canal filling material, and implants., More recently, the use of CBCT in endodontic research has enabled 3D assessment of treatments performed within the root canal system.,,,, Recent guidelines published AAE/AAOMR Joint Position Statement 2015/2016 Update recommended that limited FOV CBCT should be the imaging modality of choice when evaluating the nonhealing of previous endodontic treatment to help determine the need for further treatment, such as nonsurgical, surgical or extraction. Moreover, limited FOV CBCT was suggested for nonsurgical retreatment to assess endodontic treatment complications, such as overextended root canal obturation material, separated endodontic instruments, and localization of perforations.
CBCT units offer multiple field of views (FOVs) and voxels that can better address a variety of specific tasks. Voxel size is of paramount importance in terms of image quality and scanning and reconstruction times of CBCT images. A “voxel” describes the smallest distinguishable box-shaped part of a 3-dimensional (3D) image. In CBCT imaging, voxels are usually isotropic and range from 0.4 mm to as small as 0.075 mm.
High-resolution micro CT scanner enables three-dimensional evaluation of the root canal system after root canal filling and it is considered as gold standard for volumetric measurements. Micro CT is an innovative, non-destructive, and reproducible device that produces very thin sections and a true three-dimensional (3D) reconstruction of the object. Technically, it is possible to produce ultra-high resolutions of 1 μm ex vivo, using microfocal spot X-ray sources and high-resolution detectors, with a high radiation dose that is incompatible with human organism.,
In view of the importance of accurate determination of the exact location and volume of the residual filling material during retreatment procedures, the use of CBCT images may provide useful information for the clinician in the volumetric assessment of the amount of residual filling material in root canals. To our knowledge, no previous study assessed the effectiveness of CBCT images for the volumetric measurement of residual filling material when compared to Micro CT in root canals. Our objective was therefore, to compare three different voxel sizes of CBCT with Micro-CT images to determine residual filling material volume in root canals after retreatment procedure.
| Materials and Methods|| |
Specimen selection and preparation
The study protocol was reviewed and approved by the local ethical committee. Forty-two roots of 14 extracted human maxillary molars were used for the present study. All the samples were radiographed in a bucco-lingual direction by using intraoral photo stimulable phosphor plate and teeth with fracture, cracks, or any other defect were excluded. Debris and soft tissue remnants were cleaned and teeth were kept in 0.1% thymol solution for a week for disinfection; 24 hours before the experiment,
teeth were retrieved from thymol solution, rinsed, and stored in distilled water. After preparation of the access cavity, working length was determined by inserting a size 10 K-file (Dentsply, Maillefer, OK US). All root canals were then instrumented by using ProTaper rotary instruments (Dentsply, Maillefer, OK, US) in a crown-down manner. As the following sequence: SX file was used at 1/2 of the working length, whereas, S1, S2, F1, F2 files were used to the full working length. During shaping, each canal was irrigated between instruments with 2 mL of 2.5% sodium hypochlorite (NaOCl). Finally, the root canals were dried with paper points and obturated with a size F2 ProTaper single cone and AH Plus Sealer (Dentsply, DeTrey, Konstanz, Germany). The quality and apical extent of root canal fillings were confirmed radiographically. Teeth were stored at 37°C in 100% humidity for 2 weeks, to allow complete setting of the sealer.
Removal of root filling material
After setting of the sealer was completed, the coronal third of the root filling was removed with Gates-Glidden drills and the filling material was removed using 0.06 taper ProFile ® (Dentsply Maillefer, Baillagues, Switzerland) instruments sizes 40, 35, 30, 25, and 20 in a crown-down sequence at a speed of 600 rpm and a torque of 2.4 Ncm. After retreatment process, root canals were irrigated once again with 2 mL of 17% EDTA and flushed with 2 mL of 2.5% NaOCl. Each instrument was used to prepare a maximum of four root canals. No additional instruments and solvents were applied. A ten year experienced endodontist conducted all treatment and retreatment procedures.
Cone Beam Computed Tomography (CBCT) scanning procedures and evaluation
For CBCT scanning procedures, each tooth was placed in the appropriately prepared maxillary first molar socket of a dry skull with a natural second premolar and second molar teeth. The dry skull along with its maxilla was covered with 1.5-cm red wax as a soft tissue equivalent material. Images of each tooth inserted in the maxilla were obtained by using the ProMax 3D Max CBCT (Planmeca, Helsinki, Finland) with a flat panel sensor using low artifact reduction mode operating at 96 kVp, 1 mA, 55 × 50 mm FOV at 3 different voxel sizes 1) High resolution (0.1 mm) 2) High definition (0.15 mm) and 3) Normal resolution (0.2 mm) voxel size [Figure 1]. The exposure time and dose area product (DAP, mGy*cm 2) showed variations among different voxel sizes. The exposure time ranged between 11.948 and 12.154 seconds for 0.1-mm voxel size, 14.916 and 15.142s for 0.15-mm voxel size, and 12.042 and 12.133 seconds for 0.2-mm voxel sizes. The DAP values were 67 mGy*cm 2, 83 mGy*cm 2, and 67mGy*cm 2 for 0.1-mm, 0.15-mm, and 0.2-mm voxel sizes, respectively. Exposure time was determined by consensus. Pulpal root canal, dentin, and enamel visibility were used as indicators of optimal image quality. After CBCT data was exported as DICOM files for all the samples, volumetric measurements of residual filling material inside the canal was conducted by using 3D-DOCTOR (Able Software Corp., Lexington, MA) [Figure 2], and measurements were then compared with micro-CT image measurements. 3D-DOCTOR (Able Software Corp., Lexington, MA), is volumetric rendering software capable of vector-based segmentation technology. This software allowed segmentation of residual filling material on consecutive axial slices enabling visualization at each level apicocoronally. This ensured detailed slice-by-slice manual segmentation of the residual filling material borders using a mouse with turquoise colored delineation via trace boundary tool.
|Figure 1: a: CBCT images of residual filling materials obtained by using 0.1 mm voxel size in coronal, sagittal, and axial view. b: CBCT images of residual filling materials obtained using 0.15 mm voxel size in coronal, sagittal, and axial view. c: CBCT images of residual filling materials obtained using 0.2 mm voxel size in coronal, sagittal, and axial view|
Click here to view
|Figure 2: Volumetric measurement of residual filling material analyzed by using 3D-DOCTOR (Able Software Corp., Lexington, MA)|
Click here to view
Micro-CT scanning procedures and evaluation
Teeth were scanned postoperatively by using Skyscan 1174 micro-CT (Skyscan, Kontich, Belgium) by using following parameters: 800 (μA), 50 (kVp) and 21 μm projections within 180° rotation and 4000 (ms) scanning time for each specimen. Scan time was 60 min in duration. Three-dimensional reconstruction data was obtained with the aid of dedicated software (NRecon version 184.108.40.206, Skyscan, Kontich, Belgium) using post threshold-based segmentation and viewed by using Dataviewer Program (version 220.127.116.11, Bruker micro CT). Scanning was then transferred into CTan software (version 18.104.22.168) for image analysis. The inflection point of the micro CT absorption histogram between dentine and filling material compartments was used as segmentation threshold. Volumes of residual filling materials were calculated in mm, and the mean values for each specimen were determined. The mean percentage of residual filling material was color coded and calculated [Figure 3].
|Figure 3: Postoperative micro-CT images of residual filling materials obtained by using Skyscan 1174 micro-CT (Skyscan, Kontich, Belgium)|
Click here to view
Two observers (a three years experienced radiologist, a 10-year experienced endodontist) assessed the images obtained from 2 techniques in random order. The observers were allowed to use enhancement facilities, such as: g-curve, contrast, brightness, and zoom function if necessary. All images were evaluated separately on a 22” NEC MD213MG LCD monitor (NEC, Tokyo, Japan), at a screen resolution of 2048 × 1536 pixel and 32-bit color depth. Image sets were viewed at 1-week intervals, and repetitions performed on half of the images one month after the initial viewings. No time restriction was placed on the observers.
Mann-Whitney U test was used to compare differences between observers. Wilcoxon Test was used for the comparison of CBCT and micro CT measurements. The Spearman correlation coefficient was used to display the relationship between volume measurements at different voxel sizes of CBCT and micro CT. The significance level was set at P<.05.
| Results|| |
Mann-Whitney U test showed no statistically significant differences between the two observers for all measurements (P > 0.05). According to Wilcoxon Test, there were no statistically significant differences among different CBCT voxel sizes (0.1 mm, 0.15 mm, and 0.2 mm) in comparison to micro CT measurements for each observer (P > 0.05) [Table 1]. The Spearman correlation coefficients between CBCT at different voxel sizes significantly highly correlated with micro CT measurements (P < 0.05) [Table 2]. Furthermore, no significant differences were found between the measurements obtained by the two observers in consideration to different root canal location (P > 0.05) [Table 3]. According to micro CT measurements the average volume of residual filling material in mesiobuccal (MB), distobuccal (DB) and palatinal (P) root canals were found to be 1.456 mm 3, 1.126 mm 3, and 1.905 mm,3 respectively. Average volume of residual filling material calculated at different CBCT voxel sizes (mm) for different root canal locations and for both observers was shown in [Table 4].
|Table 1: Median, mean, and standard deviation values of the volumes of residual filling materials at different CBCT voxel sizes and micro CT measurements for each observer|
Click here to view
|Table 2: Spearman Correlations Coefficient for each voxel size and micro CT measurement for each observer|
Click here to view
|Table 3: Median, mean, and standard deviation values of the volumes of residual filling materials at different CBCT voxel sizes for each observer at different root canal locations|
Click here to view
| Discussion|| |
The success of root canal retreatment procedures depends on complete removal of obturation materials from the root canal system. Moreover, the presence of residual root canal filling material was identified as a prognostic factor related to retreatment failure that enables the effective action of instruments and irrigating solutions on debris and microorganisms responsible for the apical infection., Various methods were used to evaluate the amount of residual root filling material in vitro. One of them was splitting the roots longitudinally in order to evaluate the canal walls microscopically. Radiographs were another way to evaluate residual root filling material both in vitro and in vivo.,,, In a previous study, residual filling material was first evaluated radiographically and then by microscopy. Although, computerized morphometric analysis was regarded as an objective analysis tool, the projection of the root canal surface in either the radiographs or the digital photographs did not accurately represent the canal surface. Periapical radiography is the most common method that can be used for the evaluation of residual root canal filling material clinically. However, periapical radiographic image is the compression of three-dimensional (3-D) structures into a two-dimensional view. Therefore, they are insufficient to locate and assess residual root canal filling materials specifically in the maxillary molar region due to superimpositions of adjacent teeth and anatomical structures., The main advantage of using CBCT for endodontic applications is that it can provide three-dimensional views which intraoral radiography cannot provide with considerably higher radiation doses than 2 dimensional techniques.,, CBCT scan is an important tool for detecting intraoperative endodontic complications or difficulties, such as; complex anatomy, root resorptions, perforations, and root fractures in cases in which the diagnosis or treatment was compromised by the limitations of radiographs., We used maxillary first molar teeth in the appropriately prepared maxillary first molar socket of a dry skull with a natural second premolar and second molar teeth in order to create a more realistic clinical situation. In addition, we preferred using ProFile system for retreatment procedures. The validity and accuracy of this methodological approach has been previously demonstrated. In a recent study, authors found that by using ProFile instruments, the clinician left significantly less residual filling material when compared with ProTaper and Mtwo retreatment files.
The clinical use of different Field of View (FOVs) and voxel sizes may affect image quality along with patient radiation doses. Currently, voxel sizes between 0.1 mm and 0.2 mm were suggested and regarded as optimal for various endodontic purposes. In addition, the smallest FOV of the CBCT unit is recommended in endodontic applications due to reduced radiation dose with smaller FOV., We used the smallest FOV (55 × 50 mm) offered by the CBCT unit that is suggested for endodontic use with the advantages of lower patient exposure, shorter scan times, and smaller voxel sizes available in comparison with larger FOVs. It is possible that the use of CBCT units with a different FOV and a voxel size than the one we preferred would reveal different observer performance.
CBCT artifacts include streaks around materials as well as dark zones that affect the overall quality of the image and may cause discrepancies between the reconstructed image and the scanned object. CBCT manufacturers are now actively developing artifact-reducing algorithms to be used during image reconstruction., In the present study, we employed artifact reduction filters for both CBCT and micro CT used, however; comparison of different artifact reduction filters offered by different CBCT units was out of our scope. We preferred using low mode since endodontic filling materials cause less artifacts when compared to metal and amalgam restorations. Moreover, in applying our results to clinical situations, it should be made clear that patient motion was not a factor in this experimental in vitro set up.
Up until now, the potential of CBCT images obtained at different voxel sizes to explore the root canal system for volumetric measurement of the residual root canal filling material has not been assessed in comparison to Micro CT as the gold standard. We found that there were no significant differences among voxel sizes used (0.1 mm, 0.15 mm, and 0.2 mm) compared with Micro CT measurements. Also, pairwise comparisons showed high correlation between Micro CT and CBCT for each voxel sizes and for two observers. Volumetric measurements may vary depending on the observer performance, CBCT unit and settings and software used along with mouse sensitivity. In this study, trained researchers in using measurement tools of the software acted as observers. Furthermore, it may be speculated that differences between tooth and root canal types and curvatures may cause discrepancies between volumetric measurements. However, we found no significant differences between the measurements in consideration to different root canal location.
In a previous regenerative endodontic treatment (RET) study, volumetric measurement of a newly formed hard tissue was conducted to develop a standardized quantitative method for RET outcome analysis based on cone-beam computed tomographic (CBCT) volumetric measurements. CBCT images were obtained in high resolution, 80 μm voxel size and 6 × 6 cm FOV. Similar to our study, volumetric measurements were conducted and compared. The results showed no statistical differences and the authors found that there was a strong correlation between CBCT and micro CT volumetric measurements. Thus, they suggested that CBCT images could be used for teeth treated with regenerative endodontic procedures to detect treatment outcome and patterns of hard tissue formation for clinical relevance. Similarly, we found significant correlation between CBCT and micro-CT measurements suggesting that CBCT images have the strong potential to be used for volumetric measurement of residual root canal filling material that may be useful for retreatment procedures. However, it should be kept in mind that it is more difficult to segment unclear residual filling materials in comparison to well-defined regenerative bone tissue.
In a similar previous study, authors measured residual filling material volume by CBCT to compare the efficacy of different file systems. ProTaper retreatment files were used followed by either WaveOne reciprocating file or the Self-Adjusting File (SAF). The CBCT scans were obtained using ProMax 3D Mid (Planmeca OY, Helsinki, Finland). The samples were exposed to 90 kV and 8 mA with a FOV of 4.5 × 4.5 cm 2 and an isotropic resolution of 0.1 mm, with 12.28 s exposure time. According their results the mean postoperative volume of the root filling residue was 9.4 (±0.5) mm 3 in Group 1 and 2.6 (±0.4) mm 3 in Group 2. These residue volumes represented 26% and 7% of the original root filling volumes in Group 1 and Group 2, respectively. Analoguous to our findings, they reported that the use of two-dimensional (2D) radiographs failed to represent the real cleanliness of the canal. Therefore, CBCT with small FOV, which is readily available to more researchers than micro-CT, may be a valid tool for retreatment cases.
Authors of a previous study suggested the use of CBCT images obtained at voxel sizes ranging between 0.1 mm to 0.2 mm as an adjunct to periapical radiography for the detection of various endodontic complications. However, up until now there is no accepted protocol for the use of different voxel sizes for different endodontic tasks. Our study was the first to assess different voxel sizes for the volumetric measurement of residual filling materials in root canals. Radiation concerns regarding CBCT is a limitation of the system and its potential benefits should outweigh its hazards when prescribing CBCT. The effective dose for the CBCT unit utilized in the present study is in the range 28 to 122μSv. This is higher than the effective doses from periapical radiography taken with PSP with rectangular collimation (0.1 to 2.6 μSv). Prospective randomized clinical trials based on well-defined methodologies are needed to assess the potential benefits of CBCT in endodontic retreatment. Within the limitations of the present ex vivo study, CBCT images obtained at voxel sizes <0.2 mm appears to be an effective imaging technique for the determination of the location and volume of residual filling material in root canals for retreatment procedures.
| Conclusion|| |
All CBCT images obtained at different voxel sizes performed similarly in the detection of residual root canal filling material volume. In addition, measurements of the volumes from CBCT images highly correlated with micro-CT volume. CBCT images may provide useful information in the volumetric determination of the amount of residual filling material in root canals for retreatment procedures.
Authors declare no conflict of interest. No funding was obtained for this study. No financial support and sponsorship.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Alves FR, Marceliano-Alves MF, Sousa JC, Silveira SB, Provenzano JC Jr, Siqueira JF. Removal of root canal fillings in curved canals using either reciprocating single-or rotary multi-instrument systems and a supplementary step with the XP-endo finisher. J Endod 2016;42:1114-9.
Anjo T, Ebihara A, Takeda A, Takashina M, Sunakawa M, Suda H. Removal of two types of root canal filling material using pulsed Nd: YAG laser irradiation. Photomed Laser Surg 2004;22:470-6.
Bernardes RA, Duarte MA, Vivan RR, Alcalde MP, Vasconcelos BC, Bramante CM. Comparison of three retreatment techniques with ultrasonic activation in flattened canals using micro-computed tomography and scanning electron microscopy. Int Endod J 2016;49:890-7.
Gu LS, Ling JQ, Wei X, Huang XY Efficacy of ProTaper Universal rotary retreatment system for gutta-percha removal from root canals. Int Endod J 2008;41:288-95.
Tasdemir T, Yildirim T, Celik D. Comparative study of removal of current endodontic fillings. J Endod 2008;34:326-9.
So MV, Saran C, Magro ML, Vier-Pelisser FV, Munhoz M. Efficacy of ProTaper retreatment system in root canals filled with gutta-percha and two endodontic sealers. J Endod 2008;34:1223-5.
Unal GC, Kaya BU, Tac AG, Keçeci AD. comparison of the efficacy of conventional and new retreatment instruments to remove gutta-percha in curved root canals: An ex vivo
study. Int Endod J 2009;42:344-50.
Kfir A, Tsesis I, Yakirevich E, Matalon S, Abramovitz I. The efficacy of five techniques for removing root filling material: Microscopic versus radiographic evaluation. Int Endod J 2012;45:35-41.
Angelopoulos C, Scarfe WC, Farman AG. A comparison of maxillofacial CBCT and medical CT. Atlas Oral Maxillofac Surg Clin North Am 2012;20:1-17.
Kamburoglu K, Kolsuz E, Murat S, Eren H, Yüksel S, Paksoy CS. Assessment of buccal marginal alveolar peri-implant and periodontal defects using a cone beam CT system with and without the application of metal artefact reduction mode. Dentomaxillofac Radiol 2013;42:20130176.
Schulze R, Heil U, Gross D, Bruellmann DD, Dranischnikow E, Schwanecke U, et al.
Artefacts in CBCT: A review. Dentomaxillofac Radiol 2011;40:265-73.
Rodriguez G, Abella F, Durán-Sindreu F, Patel S, Roig M. Influence of cone-beam computed tomography in clinical decision making among specialists. J Endod 2017;43:194-9.
Rodriguez G, Patel S, Durán-Sindreu F, Roig M, Abella F. Influence of cone-beam computed tomography on endodontic retreatment strategies among general dental practitioners and endodontists. J Endod 2017;43:1433-7.
Kamburoglu K, Barenboim SF, Aritürk T, Kaffe I. Quantitative measurements obtained by micro-computed tomography and confocal laser scanning microscopy. Dentomaxillofac Radiol 2008;37:385-91.
Patel S. New dimensions in endodontic imaging: Part 2. Cone beam computed tomography. Int Endod J 2009;42:463-75.
Patel S, Dawood A, Whaites E, Pitt Ford T. New dimensions in endodontic imaging: Part 1. Conventional and alternative radiographic systems. Int Endod J 2009;42:447-62.
The American Dental Association Council on Scientific Affairs. The use of cone-beam computed tomography in dentistry. J Am Dent Assoc 2012;143:899-902.
Rödig T, Kupis J, Konietschke F, Dullin C, Drebenstedt S, Hülsmann M. Comparison of hand and rotary instruments for removing gutta-percha from previously treated curved root canals: A micro computed tomography study. Int Endod J 2014;47:173-82.
Ng YL, Mann V, Gulabivala K. A prospective study of the factors affecting outcomes of nonsurgical root canal treatment: Part 1: Periapical health. Int Endod J 2011;44:583-609.
Schirrmeister JF, Hermanns P, Meyer KM, Goetz F, Hellwig E. Detectability of residual epiphany and gutta-percha after retreatment using a dental operating microscope and radiographs an ex vivo
study. Int Endod J 2006;39:558-65.
Kamburoğlu K, Yeta EN, Yılmaz F. An ex vivo
comparison of diagnostic accuracy of cone-beam computed tomography and periapical radiography in the detection of furcal perforations. J Endod 2015;41:696-702.
Ball RL, Barbizam JV, Cohenca N. Intraoperative endodontic applications of cone beam computed tomography. J Endod 2013;39:548-57.
Yılmaz F, Koc C, Kamburoğlu K, Ocak M, Geneci F, Uzuner MB, et al.
Evaluation of 3 different retreatment techniques in maxillary molar teeth by using micro-computed tomography. J Endod 2018;44:480-4.
EzEldeen M, Van Gorp G, Van Dessel J, Vandermeulen D, Jacobs R. 3-dimensional analysis of regenerative endodontic treatment outcome. J Endod 2015;41:317-24.
Pawar AM, Thakur B, Metzger Z, Kfir A, Pawar M. The efficacy of the Self-Adjusting File versus WaveOne in removal of root filling residue that remains in oval canals after the use of ProTaper retreatment files: A cone-beam computed tomography study. J Conserv Dent 2016;19:72-6.
] [Full text]
Koç C, Sönmez G, Yılmaz F, Karahan S, Kamburoğlu K. Comparison of the accuracy of periapical radiography with CBCT taken at 3 different voxel sizes in detecting simulated endodontic complications: An ex vivo
study. Dentomaxillofac Radiol 2018;22:20170399.
Pauwels R, Beinsberger J, Collaert B, Theodorakou C, Rogers J, Walker A, et al.
Effective dose range for dental cone beam computed tomography scanners. Eur J Radiol 2012;81:267-71.
Granlund C, Thilander-Klang A, Ylhan B, Lofthang-Hansen S, Ekestubbe A. Absorbed organ and effective doses from digital intra-oral and panaromic radiography applying the ICRP 103 recommendations for effective dose estimations. Br J Radiol 2016;89:20151052.
[Figure 1], [Figure 2], [Figure 3]
[Table 1], [Table 2], [Table 3], [Table 4]